The repair of electronics is our core business, therefore, unlike original equipment manufacturers, we invest all our skill and energy in providing a high quality, cost effective repair solution geared around the needs of our customers. All our repairs and spares come with 12 months whole of unit warranty. This is why we have a number of repair options to suit your requirements:. Buying a refurbished Krohne IFC D can offer great savings over brand new and at Lektronix all our refurbished spares are covered by our comprehensive 12 month warranty, giving you complete peace of mind.
|Published (Last):||11 May 2016|
|PDF File Size:||3.62 Mb|
|ePub File Size:||9.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Installation and. Signal converters for electromagnetic flowmeters. How to use these Instructions. The flowmeters are supplied ready for operation. The primary head must be installed in the pipeline as described in the installation instructions inside the packing of the primary head.
Power the flowmeter. The system is operative. Installation Sect. Applicable to. Software Versions. Order No. Part A. System installation and start-up. Electrical connection: power su pply. Cable entries.
Connection to power. Electrical connection of separate primary head F Version. General information on signal cable A and field current cable C. Grounding of primary head. Stripping preparation of signal converter. Cable length max. Connection diagrams I and II power supply, converter and primary head. Electrical connection of outputs. Current output I. Pulse output P and status output S.
Connection diagrams for outputs and inputs. Power-on and measurement. Factory settings. Part B. IFC Operation of the si gnal converter.
Krohne operator control concept. Operating and check elements. Function of keys. Table of settable functions. Error messages in measuring mode. Examples of setting the signal converter. Description of functions. Time constant. Low-flow cutoff. Internal electronic totalizer. Pulse output P. Status output S. Entry code. Primary head. User-defined unit. Characteristic of outputs. Setting data. Part C. Special applications, functional checks, service, and order numbers. Special a pplications. HHT hand held terminal and RS adapter incl.
Stable signal outputs when measuring tube empty. Functional checks. Zero check with IFC Test of measuring range Q, Fct. Hardware information and error status, Fct. Faults and symptoms during start-up and process flow measurement.
Test of primary head. Test of signal converter. Test of signal converter using GS 8A simulator option. Cleaning the signal converter housing. Replacement of power fuse s.
Replacement of electronic unit of signal converter. Illustrations to Sect. Turning the display PCB. Retrofitting the display unit. Illustrations of the PCBs. Order numbers. Part D. Technical data, measuring principle and block diagram. Technical data. Error limits at reference conditions. IFC Signal converter. Instrument name plates. Measuring principle and function of the s ystem. Block diagram - signal converter. Part E. Printed form to accompany flowmeters returned to Krohne.
Versions IFC signal converter. Basic version standard. All operating data factory-set to your order specifications. Optionally available for operator control:. Display version option with local display and control elements. Compact flowmeter signal converter mounted direct on primary head. Signal converter in field housing, electrical connection to primary head via field power and signal cables.
Items included with supply. Software history. Hand-held HHT RS At least the same setting ranges and functional scope as preceding versions. Also, customer- and application-specific add-on equipment possible, which has to be installed and activated by the factory.
Documented by enclosures to these Installation and Operating Instructions. Please note: Connect HHT only to devices without display and operator control software. Does not contain the functional scope of the currently valid standard version; this has been taken into account in the documentation of the respective national language. System description. Electromagnetic flowmeters with the IFC signal converter are precision instruments designed for linear flow measurement of liquid products.
This is equivalent to a flow velocity of 0. Product liability and warranty.
KROHNE IFC 010
Display shows. Remedial action. Delete error message in. Note: no counting during. Current output overranged.
KROHNE IFC 010 Converter EN User Manual
Be the first to write a review. Skip to main content. Email to friends Share on Facebook - opens in a new window or tab Share on Twitter - opens in a new window or tab Share on Pinterest - opens in a new window or tab. Add to Watchlist. People who viewed this item also viewed. Picture Information.